The influence of external factors on the accuracy of non-invasive measuring of oxygen in blood
AbstractIn this paper we investigated a pulse oximetry-based method for mobile devices. This method obtains bio-signals related to blood pulsation in transparent parts of body. The most widely accepted field for use of this method is hospital care. In these cases a pulse oximeter is the best solution for the monitoring of emergency patients. A promising field for pulse oximetry is physical exercise. It only requires simple clips such as ear-clips, finger-clips, headbands etc. However this method presents some difficulties: weak signal, noise ratio, motion artefacts, low perfusion. We used a MAX30100 Oximeter and Heart Rate Sensor integrated circuit to obtain signals of blood pulse waves from red and infrared light emission diodes (LED). This device measures the oxygen saturation of a person’s blood by placing an LED and a photodetector against the thin skin of a person’s body, such as a fingertip, wrist or earlobe. The MAX30100 is a 14-pin surface mount integrated circuit that contains sensors for measuring a person’s heart rate. It can also indirectly determine the oxygen saturation of a person’s blood. The MAX30100 provides a complete pulse oximetry and heart rate measurement solution for medical monitors and wearable fitness devices. As each LED emits light into a person’s finger, the integrated photodetector measures variations in light caused by changes in blood volume. An integrated 16-bit analog to digital converter (ADC) with programmable sample rate converts the photodetector output to a digital value. The MAX30100 filters out ambient light that can interfere with an accurate reading. Data are read through a serial I2C interface to computer for further processing. The LED current can be programmed from 0 to 50 mA with proper supply voltage. The LED pulse width can be programmed from 200 µs to 1.6 ms to optimize measurement accuracy and power consumption based on use cases. The SpO2 algorithm is relatively insensitive to the wavelength of the infrared LED, but the red LED’s wavelength is critical to correct interpretation of the data. The temperature sensor data can be used to compensate errors with ambient temperature changes. During the experiments we registered oxygen concentration values under different conditions. With low muscular activity we observed high stability and repeatability of measuring values under various exterior conditions. However, with high muscular activity there were various artefacts in the gauged signals that led to contortion of effects. We identified the boundaries of the validity of measuring and propose the use of an adapted filter in order to distinguish pulse waves from optical signals more reliably. These devices can be applied in fitness training , medical monitoring and used as wearable devices.
Barata, D., Louzada, G., Carreiro, A., Damasceno, A., 2013. System of acquisition, transmission, storage and visualization of pulse oximeter and ECG data using android and MQTT. Procedia Technology 9, 1265–1272. >> doi.org/10.1016/j.protcy.2013.12.141
Batchelder, P.B., Raley, D.M., 2007. Maximizing the laboratory setting for test devices and understanding statistical output in pulse oximeter. Anesth. Analgю 105, 585–594. >> doi.org/10.1213/01.ane.0000268495.35207.ab
Baura, G.D., 2012. Chapter 11 – Pulse Oximeters. In: Medical Device Technologies. Academic Press, Oxford, 237–255. >> doi.org/10.1016/b978-0-12-374976-5.00011-6
Davila, D.G., Richards, K.C., Marshall. B.L., O’Sullivan, P.S., Gregory, T.G., Hernandez, V.J., Rice, S.I., 2002. Oximeter performance: The influence of acquisition parameters. Chest 122(5), 1654–1660.
Dixit, A., Sharma, R., Barai, S., 2015. Developing and prototyping pulse oximeter for elderly people. Materials Today: Proceedings 2(4–5), 1560–1567. >> doi.org/10.1016/j.matpr.2015.07.082
Edrich, T., Flaig, M., Knitza, R., Rall, G., 2000. Pulse oximetry: An improved in vitro model that reduces blood flow-related artifacts. IEEE T. Bio-Med. Eng. 47(3), 338–343. >> doi.org/10.1109/10.827294
Evstigneev, A.R., 1999. Lazernaja biofotometrija: Pribory, rezul’taty i perspektivy primenenija [Laser biophotometry: Devices, results and perspectivs]. Optojelektronika i Biomedicina. JePP MZMP RF, Moskow, 125–142 (in Russian).
Grote, L., Sommermeyer, D., Zou, D., Eder, D.N., Hedner, J., 2011. Oximeter-based autonomic state indicator algorithm for cardiovascular risk assessment. Chest 139(2), 253–259. >> doi.org/10.1378/chest.09-3029
Kreps, E.G., 1978. Oksigemometrija [Oxihemometry]. Medicinа, Moscow (in Russian).
Levene, S., Lear, G.H., McKenzie, S.A., 1989. Comparison of pulse oximeters in healthy sleeping infants. Respir. Med. 83(3), 233–235. >> doi.org/10.1016/s0954-6111(89)80037-x
Mendelson, Y., 1992. Pulse oximetry: Theory and applications for noninvasive monitoring. Clinical Chemistry 38, 1801–1807 (in Russian).
Moyle, T.B., 1994. Pulse Oximetry. BMJ books. 35–36.
Payne, J.P., Servinghaus, J.W., 1986. Pulse oximetry. Springer, 21–22. >> doi.org/10.1007/978-1-4471-1423-9
Perutz, M.F., 1978. Hemoglobin structure and respiratory transport. Sci. Am. 239(6), 92–125. >> doi.org/10.1038/scientificamerican1278-92
Pole, Y., 2002. Evolution of the pulse oximeter. International Congress Series 1242, 137–144. >> doi.org/10.1016/s0531-5131(02)00803-8
Rheineck-Leyssius, A.T., Kalkman, C.J., 1999. Advanced pulse oximeter signal processing technology compared to simple averaging effect on frequency of alarms in the operating room. J. Clin. Anesth. 11(3), 192–195. >> doi.org/10.1016/s0952-8180(99)00034-3
Rogatkin, D.A., Lapaeva, L.G., 2003. Perspektivy razvitija neinvazivnoj spektrofotometricheskoj diagnostiki v medicine [Prospects for the development of non-invasive diagnostic medical spectrophotometry]. Medicinskaja Tehnika 4, 31–36 (in Russian).
Ross, E.M., Matteucci, M.J., Shepherd, M., Barker, M., Orr, L., 2013. Measuring arterial oxygenation in a high altitude field environment: Comparing portable pulse oximetry with blood gas analysis. Wilderness Environ. Med. 24(2), 112–117. >> doi.org/10.1016/j.wem.2012.11.009
Shah, N., Ragaswamy, H.B., Govindugari, K., Estanol, L., 2012. Performance of three new-generation pulse oximeters during motion and low perfusion in volunteers. J. Clin. Anesth. 24(5), 385–391. >> doi.org/10.1016/j.jclinane.2011.10.012
Sola, J., Chetelat, O., 2007. Combination of multiple paths in pulse oximetry: The finger ring example. Proceedings of the 29th Ann. Int. Conference of IEEE EMBS. 6697–6698. >> doi.org/10.1109/iembs.2007.4353897
Trivedi, N.S., Ghouri, A.F., Shah, N.K., Lai, E., Barker, S.J., 1997. Effects of motion, ambient light, and hypoperfusion on pulse oximeter function. J. Clin. Anesth. 9(3), 179–183. >> doi.org/10.1016/s0952-8180(97)00039-1
Wieben, G.O., 1997. Light absorbance in pulse oximetry, in design of pulse oximeter. Webster J. IOP Publishing, 40–55. >> doi.org/10.1887/0750304677/b470c4
Authors retain copyright and grant the journal right of first publication with the work simultaneously licensed under a Creative Commons «Attribution» 4.0 License that allows others to share the work with an acknowledgement of the work's authorship and initial publication in this journal.